- HOME Email Webmaster
Related Links :
- Graphite Furnace AAS
- Spectral interferences
- Ultraviolet spectroscopy
- Hg/cold vapor and As/hydride
- Atomic absorption spectroscopy
Basic IR Spectroscopy
Intfrared spectroscopy deals with the infrared region of the Electromagnetic Spectrum. Only in this region is the energy intense enough to exite bonds in a molecule to absorb energy. Molecules absorb these frequencies of light because they correspond to frequencies of vibrations of bonds in the molecule.
Regions of the IR spectrum
4000 cm1 – 1300 cm1 Absorbance bands in this region can indicate what functional groups are present in an unknown compound.
1300 cm1 – 400 cm1 Absorbance bands in this region are unique to each molecule, like a fingerprint! Spectra are compared to known spectra.
The spectrophotometer detects this absorption and records it as a plot of transmission versus frequency.
Since each type of bond vibrates at a characteristic wavenumber , infrared spectroscopy has become a useful technique to identify functional groups in a molecule. The method involves the examination of the twisting, bending, rotating, vibrational motions of atoms and molecules.
Dispersive Infrared Spectrophotometer
This is a double beam scanning IR instrument that makes use of a diffraction grating in a monochromator to disperse the different wavelengths of light.
The key components are the light source, entrance slit, grating and detector. Radiation from the source is directed along both sample and reference and then into the diffraction grating. The grating disperses the light into spatially separated wavelengths which are selectively directed by a narrow slit towards the detector.
The wavelengths are measured one at a time, with the slit controlling the bandwidth. The grating and slit combination selects the wavelength being measured, and the detector produces a spectrum of a plot of transmittance against frequency.
Fourier Transform SpectrophotometerThe FTIR instrument relies upon interferences of various frequencies of light to produce a spectrum. It has a source, sample, two mirrors, a laser reference, and detector, but the assembly of components also include a beamsplitter and the two strategic mirrors that function as an interferometer.
The source energy strikes the beamsplitter and produces two beams of roughly the same intensity. One beam strikes the fixed mirror and returns to the beamsplitter. The other beam goes to the moving mirror. The motion of the moving mirror makes the total pathlength variable versus that taken by the fixed mirror beam. When these two beams meet up again at the beamsplitter, they recombine, and the difference in their path lengths create constructive and destructive interference, an interferogram.
The recombined beam passes through the sample. The sample absorbs all the wavelengths characteristic of the its spectrum and then subtracts specific wavelengths from the interferogram. The detector now reports variation in energy-versus-time for all wavelengths simultaneously. A laser beam is superimposed to provide a reference for the operation of the instrument.
A mathematical function known as a Fourier transform is used to convert the intensity-versus-time spectrum into an intensity-versus-frequency spectrum.
Components that make up the IR spectrophotometer
Radiation source Monochromator Detector Sample NERST glower
The radiation source is composed of an inert solid which is electrically heated to a temperature in the range 1500-2000 K. The heated material will then emit infra red radiation.
The Nernst glower
This is a cylinder (1-2 mm diameter, approximately 20 mm long) of rare earth oxides. Platinum wires are sealed to the ends, and a current passed through the cylinder. The Nernst glower can reach temperatures of 2200 K.
The Globar source
This is a silicon carbide rod (5mm diameter, 50mm long) which is electrically heated to about 1500 K. Water cooling of the electrical contacts is needed to prevent arcing. The spectral output is comparable with the Nernst glower, execept at short wavelengths (less than 5 m) where it's output becomes larger.
The incandescent wire source
This is a tightly wound coil of nichrome wire, electrically heated to 1100 K. It produces a lower intensity of radiation than the Nernst or Globar sources, but has a longer working life.
Detector Type Measurement Construction Spectral Region cm-1 Bolometer Resistance Pt : Ni 4000 - 400cm-1 Thermocouple Thermoelectric effect Bi : Sb 4000 -400cm-1 Golay Thermal expansion Pneumatic cell 4000 - 400cm-1 Pyroelectric Capicitance Triglycerine SO4 Mid range Photoconductor
10000 - 2777cm-1
6666 - 1724cm-1
5000 - 1666cm-1
5000 - 525cm-1
Reaction photons InGaAs
24287 - 5882cm-1
10000 - 3226cm-1
10000 - 1818cm-1
ThermocouplesConsist of a pair of junctions of different metals; for example, two pieces of bismuth fused to either end of a piece of antimony. The potential difference (voltage) between the junctions changes according to the difference in temperature between the junctions
Golay cellThis consist of a glass-filled enclosure with a diaphragm that expands on IR radiation and gives a change in signal due to the amount of movement of the diaphragm.
Pyroelectric detectorsThis is made from a single crystalline wafer of triglycerine sulphate sandwiched between two electrodes. When an electric field is applied and the field is removed, the polarization persists, it is temperature dependent and the semi-conductor acts as a capacitor. The incident infrared heating causes a change in the capacitance of the material.
The pyroelectric effect depends on the rate of change of the detector temperature and not on the temperature itself, which allows for faster response time.
Pyroelectric detectors are used in mid-infrared region, works at room temperature and applied in most FT-IR instruments..
Photoconductive DetectorsPhotoconductive detectors are thin-film semi-conductor devices which undergo a change in conductivity when exposed to varying quantities of infrared radiation. This change results in an increase in the amount of current flowing through the device, and the output of the detector is either a change in detector current or a change in voltage developed across a load resistor.
The PbS and PbSe photoconductive detectors make use of the photoconductive effect that resistance is reduced when infrared radiation enters the detecting material.
The incoming light produces free electrons which can carry electrical current so that the electrical conductivity of the detector material changes as a function of the incident light and a change in detector current. Example is the PbS and PbSe photoconductive detectors.
Photovoltaic DetectorThis detector contains a junction in a semi-conductor material between a region where the conductivity is due to electrons, and a region where the conductivity is due to holes called a p-n junction. A voltage is generated when IR radiation strikes the device and the output is a change in voltage. Example is the InGaAs fabricated detector.
The energy gap of the InGaAs pin photodiode will vary according to the composition ratio of the In and Ga in the detecting elements. Various IR detectors with different spectral response ranges can be manufactured by altering the composition ratio.
Photoelectric detectorsThe mercury cadmium telluride detector comprise a film of semiconducting material deposited on a glass surface, sealed in an evacuated envelope. Absorption of IR promotes non-conducting valence electrons to a higher, conducting, state. The electrical resistance of the semiconductor decreases. These detectors have better response characteristics than pyroelectric detectors and are used in FT-IR instruments - particularly in combination with other instruments such as GC – FT-IR.
Sample and Sampling Techniques
Sample Type Sample cell Technique Procedure Solid NaCl Mull Sample slurry with nujol
Sample is crushed in agate mortar and pestle and mixed and ground together with nujol to a thick paste. A spot is applied between two NaCl plates and mounted in path of IR beam of instrument and a spectrum is recorded.
Solid KBr Pressed pellet Sample pressed in Die
Dried sample film The sample is dissolved in a solvent which is non-aqueous, non-reacting, non IR absorbing in the range to be measured. A drop of this solution is placed on NaCl plate and the solvent evaporated to dryness leaving a thin sample film.
Solid NaCl plates Thin film Sample dries on plate
Pressed pellet One milligram sample to 100 milligram KBr is finely ground and compressed into a transparent disc using an evacuable die and hydraulic press. Another disc is similarly made with pure dry KBr which is used as reference.
Solid ATR crystal Evanescent wave Direct into IR beam
ATR Evanescent wave This is an ATR crystal accessory used with IR spectrophotometer which allows both solid and liquid samples to be analyzed in their natural state without further sample preparation. Liquid samples are simply poured over the crystal and solid samples are clamped onto it. IR lightis passed through it, an evanescent wave is formed which is collected and measured by the detector.
Liquid NaCl Liquid cell Neat sample Fill Liquid cell
1. Liquid sample cell Cell consist of two high quality pressed NaCl windows sealed and separated by a thin Teflon gasket and wetted with mercury. Cell thickness for most liquids is usually 0.01 – 0.05mm.
Gas NaCl or KBr gas cell Gas expansion Fill evacuated cell